This product is used in combination with the AW channel cleaning adapter, leak tester and
chemical tank to provide the following functions.
This product can be combined with the AW channel cleaning adapter to drain water from
the OES scope and EVIS scope air and water channel.
This product can be combined with the leak tester to test OES scopes and EVIS scopes
for water leaks by applying air pressure to the interior of the scopes and then submerging them in water to look for an air bubbles.
This product can be combined with the AW channel cleaning adapter and chemical tank to feed rubbing alcohol into the OES scope and EVIS scope air and water channel in order to dry them
This product simplifies draining water from the OES scope and EVIS scope air and water pipelines. (AW channel cleaning adapter)
This product simplifies the water leak tests for the OES scope and EVIS scope. (Using the leak tester.)
The product can be installed on a wall near a sink with the included hanger.
This product can be installed on the left or right side of a sink.
This product simplifies feeding rubbing alcohol into the OES scope and EVIS scope air and water channel in order to dry them. (AW tube cleaning adapter, chemical tank)
OES scopes and EVIS scopes can be used.
Unless the AW channel cleaning adapter, leak tester and chemical tank are used, draining water from the air and water channel, leak inspections and feeding rubbing alcohol for drying the OES scope and EVIS scope are not possible.
In order to drain the water feeding channel, cover the water supply connector with a finger. (Same as with the former model.)
The ambient operation conditions are as follows:
a) Temperature: 10 to 40 ºC
b) Humidity: 30 to 85%
c) Atmospheric pressure: 700 to 1060 hPa
d) Not to be used in an atmosphere with inflammable gases.
The power supply conditions are as follows:
a) Rated power supply voltage: 100 V, 120 V, 220 to 240 V (according to the intended market)
b) Rated input current: 0.4 A, 0.4 A, 0.2 A
c) Frequency: 50/60 Hz
d) Allowed voltage fluctuation: ±10%
4-1 Main Unit Specifications
4-1-1 Applicable Endoscopes
4-1-2 Pump Used
Electromagnetic vibrator pump
4-1-3 Air Feeding Performance
At the scope socket: 4 2.84 x 10 Pa or greater at 0 ml/min; 4 1.76 x 10 Pa or greater at 2000 ml/min;
Note: Measured at the rated power supply voltage and a room temperature of 20 to 25 ºC
4-1-4 Control Method
By turning the power switch ON/OFF.
4-1-5 Hook Installation
Can be hung down by connecting the hanger to the hook installed in a wall.
(The hook is screwed into a wall and fixed.)
4-1-6 External Dimensions
Width 85 x Height 157 x Depth 170 mm
4-1-8 Power Supply
Voltage: 100 V, 120 V, 220 to 240 V AC
Frequency: Both 50/60 Hz
Current: 0.4A (100 V, 120 V), 0.2A (220 V to 240 V)
Voltage fluctuation rate:within ±10%
4-2 Ambient Environment
4-2-1 Ambient Temperature
10 to 40ºC
4-2-2 Relative Humidifty
30 to 85%
4-2-3 Atmospheric Pressure
700 to 1060 Pa
4-2-4 Inflammable Atmosphere
Use in an inflammable atmosphere is prohibited.